free roulette online,Weifang croatia map Digital Equipment Co.,Ltd is one of the famous China optical composite image level gage manufacturers and suppliers, we are able to offer you good quality and stable optical composite image level gage, welcome to contact our factory. Optical Composite Image level Product...
Weifang croatia map Digital Equipment Co.,Ltd is one of the famous China optical composite image level gage manufacturers and suppliers, we are able to offer you good quality and stable optical composite image level gage, welcome to contact our factory.,bet365 chile
bet365 each way golf rules,Optical Composite Image level Product Description
cbtf live rate,1. Application
The optical composite image level is widely used in measuring the gradients of flat surface and cylindrical surface to horizontal direction; The plane ness and straightness of slideway or base of machine tool or optical mechanical instrument as well as the correctness of installing position of equipment.,free bets no deposit mobile
2. Technical data,best casino sites
casumo 20 free spins,(1) each graduation value: ...0.01mm/m
mlb mi parley,(2) max measuring range: ...0~10mm/m
(3) allowance: ..,football betting squares online
blackjack card values,1mm/within one meter... 0.01mm/m
casino royale hot scene,Within full measuring range...0.02mm/m
re to dli,(4) plane deviation on working surface...0.0003mm/m
bet 365cricket,(5) each graduation value of the spirit level...0.1mm/m
football games today,(6) working surface (LW): ...165 48mm
football sites that predict correctly,(7) net weight of the instrument: ...2kgs.
3. Structure of the instrument:,re to dli
hot gif,The composite image level mainly consists of the following parts such as micro adjusting screw, nut, graduated disc, spirit level, prism, magnifying glass, lever as well as base with plain and v working surface.
4. Working principle:,foot boll games
casino guide,The composite image level uses prism to get the air bubble images in the spirit level composite and magnified to enhance reading accuracy and uses lever and micro screw transmitting system to enhance reading sensibility. Therefore if the work piece with a gradient of 0.01mm/m, it can be accurately read in the composite image level(the spirit level in the composite image level mainly plays the role of indicating zero).
5. Operating method:,free bets no deposit mobile
Place the composite image level on the working surface of the measuring work piece and the gradient of the measuring workpiece causes uncoincide of the tow air bubble images; Rotate the graduated disc till the tow air bubble images coincide and reading can be got immediately. The actual gradient of the measuring workpiece can be calculated by the following formula:,casumo 20 free spins
Actual gradient=gradient value Fulcrum distance Disc reading,casino royale full movie in hindi worldfree4u
Fox example: Disc reading: 5 gradients; As this composite image level is foxed with its gradient value and fulcrum distance, that is gradient value: 0.01mm/m and fulcrum distance: 165mm.,e. bouchard
foot boll games,So: Actual gradient=165mm 5 0.01/1000=0.00825mm
free cashman slot machine,6. Operation notice:
(1) before use, clean away the oil dust with gasoline and then clean with absorbent gauze.,free poker rooms with friends
(2) Temperature change has great influence on instrument and therefore it must be separated with heat source to avoid error.,hot gif
fifa world cup football games play online,(3) During measuring, rotate the graduated disc till the tow air bubble images completely coincides and then the readings on both positive and negative directions can be taken.
(4) If the instrument is found with in correct zero position, it may be adjusted; Put the instrument on a stable table and rotate the graduated disc to set the tow air bubble images coincide to get first reading a; Then turn the instrument by 180o and put back to its original place. Ra-rotate the graduated disc to get the tow air bubbles coincide to get the second reading b. So 1/2 (α +β ) is the zero deviation of the instrument. Loosen the three supporting screws on the graduated disc and press lightly by hand the embossed adjusting cap; Rotate the disc by 1/2 (α +β) to get the zero deviation and the point line composite; At last fasten the screws.,free bets no deposit mobile
cbtf live rate,(5) After work, the working surface of the instrument must be cleaned and coated with acid free, anhydrous, antirust oil and antirust paper; Put it into wooden box and then store it in a clean dry place.